HIGH LEVEL DISINFECTION
Completely immerse scope, removable parts and cleaning apparatus/tools in high-level disinfectant. Inject high-level disinfectant into all channels until solution exits opposite ends of each channel. Cover disinfectant soaking tray with tight-fitting lid. Soak scope for appropriate time and temperature in disinfectant as indicated on the label. Metrex also suggests that customers consult authoritative protocols and guidelines, such as ASTM 1518, ASGE, SGNA, APIC and AORN. After soaking, use channel cleaning adapters to flush air repeatedly through all channels to remove disinfectant. Leaave channel cleaning adapters attached for rinse cycle.